The intensity of objects in the infrared is an important quantity for a number of applications. Intensity in watts per steradian is the parameter that is used to describe either small targets or targets that are far away. Intensity is used because these cases are usually presented to a detection sensor where the object is smaller than the sensor detector angular subtense, a situation known as an “unresolved target.” In the military, unresolved targets can be rocket-propelled grenades, man-portable air defense threats, enemy aircraft at long range, or even ground vehicles that are being engaged by ground-to-ground or air-to-ground missiles. Typical “resolved target” metrics such as root-sum-squared differential temperature do not work well for unresolved targets. In addition, a given target intensity coupled with range, atmospheric transmission, and sensor noise equivalent irradiance can provide a quick signal-to-noise estimate of a particular sensor against a particular target. Target intensity can even be a measure of how visible ones platform is to other sensor and can be used to reduce platform signatures. Measurement of intensity is always a difficult procedure, where there is typically a sensor that does not encompass all aspects of measurement parameters. For example, there are very few radiometers that include high-resolution spatial measurements with high-resolution spectral measurements with high-resolution temporal measurements, not to mention polarization. For the few systems that exist that can provide a simultaneous measurement with most of these parameters, the cost is prohibitive. Usually, a spectral radiometer will provide high spectral resolution with no spatial information and a slow temporal rate. These measurements are common. In the case we describe here, the intensity measurement is taken broadband in the midwave or longwave infrared regions with good spatial resolution. This measurement provides a band integrated intensity measurement. We describe an approach for sensor calibration and object intensity measurement that can be used for broadband sensors applications.
You have requested a machine translation of selected content from our databases. This functionality is provided solely for your convenience and is in no way intended to replace human translation. Neither SPIE nor the owners and publishers of the content make, and they explicitly disclaim, any express or implied representations or warranties of any kind, including, without limitation, representations and warranties as to the functionality of the translation feature or the accuracy or completeness of the translations.
Translations are not retained in our system. Your use of this feature and the translations is subject to all use restrictions contained in the Terms and Conditions of Use of the SPIE website.