Silicon-On- Insulator (SOI) technology has huge potential in fabricating compact devices for various applications such as integrated optic waveguides, directional couplers, resonators etc. In this work, we present the analysis of a biosensor based on an integrated optic racetrack resonator, interrogated by a bus waveguide. The biomaterial is applied as a cladding layer. Here we analyze the coupling between the resonator and the bus waveguide, and its dependence on the bio layer. In traditional analysis, the effective refractive index and resonator total path length are the factors influencing the resonant wavelength. Our analysis shows that all parametric values decrease with increase in waveguide width and spacing. The inclusion of waveguide mode overlap and perturbation in coupled mode equation results in enhanced resonator sensitivity of an order of magnitude
We report the simulation and analytical results obtained for homogenous or bulk sensing of protein on Siliconon-
insulator strip waveguide based microring resonator. The radii of the rings considered are 5 μm and 20 μm;
the waveguide dimensions are 300 × 300 nm. A gap of (i) 200 nm and (ii) 300 nm exists between the ring and
the bus waveguide. The biomaterial is uniformly distributed over a thickness which exceeds the evanescent field
penetration depth of 150 nm. The sensitivities of the resonators are 32.5 nm/RIU and 17.5 nm/RIU (RIU - Refractive index unit) respectively.
In this paper we propose and analyze a novel racetrack resonator based vibration sensor for inertial grade application.
The resonator is formed with an Anti Resonance Reflecting Optical Waveguide (ARROW) structure which offers the
advantage of low loss and single mode propagation. The waveguide is designed to operate at 1310nm and TM mode of propagation since the Photo-elastic co-efficient is larger than TE mode in a SiO2/ Si3N4/ SiO2. The longer side of the resonator is placed over a cantilever beam with a proof mass. A single bus waveguide is coupled to the resonator structure. When the beam vibrates the resonator arm at the foot of the cantilever experiences maximum stress. Due to opto-mechanical coupling the effective refractive index of the resonator changes hence the resonance wavelength shifts. The non uniform cantilever beam has a dimension of 1.75mm X 0.45mm X 0.020mm and the proof mass has a dimension of 3mm X 3mm X 0.380mm. The proof mass lowers the natural frequency of vibration to 410Hz, hence designed for inertial navigation application. The operating band of frequency is from DC to 100Hz and acceleration of less than 1g. The resonator has a Free Spectral Range (FSR) of 893pm and produces a phase change of 22.4mrad/g.